I don’t know how to do that. It’s generally not recommended since it can make it less clear what’s happening, but it might explain what you should look for if you don’t see a character added in existing code. That sounds like an issue with the USB adapter and not the Instrument. If this is not there, you probably need to manually append the termination character to the string that you wire to VISA Write. Does anyone have any pointers? I have tried that it it does not work.
|Date Added:||26 August 2009|
|File Size:||11.90 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I have tried that it it does not work.
Benchlink data logger 3 download
Auto-suggest helps you quickly narrow agilent benchlink your search results by suggesting possible matches as you type. Make sure to send the appropriate termination character – this might be specified in the initialization VI, in which case you’ll see a property node labelled ASLR End Out. It might be the same command set for an FTDI device agilent benchlink which you must send:. That sounds like an issue with the USB adapter and not the Instrument.
I’m able to take readings and communicate with the instrument over Ethernet, but the USB connection just produces the error “Not able to execute command in benchlinl mode”.
The error is errorwhich the manual indicates is:.
Again, the existing VIs should give some guidance. If so, check here for details: Message 5 of Does anyone have any pointers? See page 56 of the instruments manual. Help the Community and future reviewers by marking agilent benchlink as follows: The BenchLink software probably sends a command to the adapter to be the controller in charge; otherwise, it acts just like a bus listener.
We’re using an adapter to convert the GPIB connection in both cases. If this is not there, you probably need to manually append the termination character to the string that you wire to VISA Write. I’ve looked through the manual for the device and tried some google-fu but can’t agilent benchlink it out.
I’m working on a piece of LabView software agilent benchlink and can’t seem to take readings from the Agilent Data Logger we’re working with when we use Ahilent to connect to the agklent.
What happens with the USB is when the read command is executed in the lab-view program the Agilent beeps and then displays that there is an error. It’s generally not recommended since it can make it less clear what’s happening, but it might explain what you should look for if you don’t see a character added in existing code.
BenchLink Data Logger 3
Message 1 of Message 4 of I don’t recall if you need a leading: Message 10 of Agilwnt is I think the command you want. Message agilent benchlink of I’m having issues with finding what command I need to send from the remote interface. Agilent “Not able to execute command in local mode” when operating over USB adapter.
Also, make sure that the Ethernet and USB connections are not given the same name. It sounds agilent benchlink there should be a pretty simple fix – I just need to take the device out of local mode – but It might be the same command set for an FTDI device in which you must send: Did you check if the instruments USB interface is enabled?
Free benchlink data logger 3 download (Windows)
Agilent benchlink USB adapter is an Agilent b. Message 8 of The error is errorwhich the manual indicates is: I looked through the VIs in the instrument driver and couldn’t find anything related to toggling the mode of the device. Message 2 of